SURECOM Colour Graphic ANTENNA ANALYZER SA-160 0-60MHz
SA-160 antenna analyzers designed for testing, checking, tuning or repairing antennas and antenna feed lines.
■ 2.2 ” precision LCD screen,High-lumen LCD brightness, easy to read in direct sunlight
■ Heavy duty Aluminum enclosure,strong anti-interference ability.
■ Very compact and convenient for field use
■ Built-in rechargeable Li-ion battery
■ Full Band control (No need select Band )
■ Easy setup, only four control buttons with convenient on-screen soft menus
■ Constant Battery Status Indicator and Auto Shut-off
■ Built-in timer shows elapsed time when unit is turned on
■ High efficiency IC, low power consumption, the longest working hours
■ More choice measurement frequency Span .
■ One botton quick scan mode HF1,HF2,HF3 or 3 band by full screen.
■ Auto marker the Best s.w.r. level reference
|Power Source:||Size 14500 of Lithium battery.|
|Charge Controller:||Smart charger, LED charging-status indicator|
|Voltage Monitor:||On-screen DVM, plus battery condition icon|
|Power Savers:||Auto-off timer, Auto Run/Stop RFgenerator control|
|RF Connector:||BNC -female|
|Control and Display|
|LCD:||2.2 inch high-output color TFT|
|Function Keys :||4 botton, with on-screen (soft-menu) identifiers|
|Tuning:||Rotary encoder with press-in tuning-step selection|
|Weight and Dimensions|
|Size:||7 cm wide x12cm height H x 2.5 cm|
|Weight:||6.1 oz (0.172 kg)|
Turn on and off the power
Turn on the power, Press and hold White botton when the dot disappear
from left up angle ,. top Menu appeared on the screen.
Turn off the power, You have 2 method :
a)On Top Menu page ,Press and hold the White [F1 ] botton at 5 second .
b)On Top Menu page , Rotary encoder on "Power off", Press the [F2 Enter]
You can turn the knob select you want function and press [F2.Enter] botton .
2.0 ANTENNA TEST EXAMPLES by Scan mode
2.1[Scan] mode page
In the Top Menu screen: Rotary encoder on "Scan", Press the [F2 Enter], to frequency mode measurement button, enter Scan (scan mode) screen.
Press [F.2 scan ],button to start the scan operation. During scanning, in order to ensure measurement accuracy, each frequency will stay for short while. one scanning cycle will take about 1-2 seconds. Once SCAN is done, it will enter ”Present” mode (result) screen
You have results on the screen : Show the SWR curve only (because default setting SWR curve only , Zx curve, Rs curve Disable) You can Enable other curve on the screen: On Top Menu page,
Rotary encoder on "Zx", Press the [F2 Enter] to enable . Other cure will Disable when Power Off , If want enable in the future, Rotary encoder on "Save", Press the [F2 Enter] to save setting .
2.2 1. lnfull (default SCAN range 0.56MHz -59.9MHz,span 62.1MHz )
* Long press to [SCAN ]and press [F3 UPDn] select "1.Infull",and then release the scan botton.
2.lrnMk=(Learn marker move to center frequency)
* Long press to [SCAN ]and press [F3 UPDn] select "2.IrnMk",and then release the scan botton.
view the scan frequency point on the curve on each scan of the specific parameters of the measurement results.
- Press the [F3.View ] button to start the view operation mode. rotary encoder ,the white triangle curve marker is move
- Dot scale will display on screen, Press bottom change the Dot scale,You have 3 options of Dot scale mode for choose,[Dot =x1] [Dot=x10][Dot= x25] . (230KHz,2.3MHz,5.8MHz)
SWR Full scale will display on screen,Press bottom change the graphically plot SWR,You have 3 options scale mode for choose, [SWR=x2 ][SWR= x 0.5][SWR=x1] .(1-9,1-19,1-4.5)
2.4[ Quick 1] start Scan mode:
Quick view the scan frequency point on the curve on each scan
Press the button [Quick 1 ] to start the Quick 1 operation mode .You have 3 options scan mode for choose .
1)Press the button [F1. Scan 1 ] 0.5 to 27.3MHz can of the specific parameters of the measurement results.
2)Press the button [F2. Scan 2 ] 16.6 to 43.4MHz can of the specific parameters of the measurement results.
3)Press the button[F3. Scan 3 ] 33.6 to 60.0MHz can of the specific parameters of the measurement results.
1 x SURECOM SA160 (ENGLISH VERSION) Colour Graphic Antenna Analyzer
1 x English User"s Manual
1 x Mirco Usb Charger Cable
1 x AC Adapter 100-240V (Depends on your country)